### Abstract

We have determined the complex uniaxial dielectric tensor of bulk rutile titanium dioxide (110), (100) and (111) samples using reflection generalized ellipsometry, which measures both the diagonal and off-diagonal elements of the reflection Jones matrix. Data were acquired using three commercially available ellipsometers, each covering the following spectral ranges: 0.148 to 0.292, 0.200 to 1.7 μm; and 1.7 to 33 μm. Generalized ellipsometry measures three complex ratios involving all four Jones matrix elements. In principle, this means that the complex dielectric tensor of a uniaxial crystal can be determined in a single measurement, provided that the sample is oriented such that the off-diagonal components of the Jones matrix are non-zero. To improve our results, we measure the samples at several rotational orientations around the surface normal. This insures that the probing electric fields vibrate along substantially different directions with respect to the optic axis. In some cases, we also varied the angle of incidence. The dielectric tensor was determined at every wavelength directly from a simultaneous fit to data from all rotational orientations and incident angles. A similar methodology should be applicable to a wide range of anisotropic optical materials.

Original language | English (US) |
---|---|

Pages (from-to) | 19-29 |

Number of pages | 11 |

Journal | Proceedings of SPIE - The International Society for Optical Engineering |

Volume | 4103 |

State | Published - Jan 1 2000 |

Event | Optical Diagnostic Methods for Inorganic Materials II - San Diego, USA Duration: Aug 3 2000 → Aug 4 2000 |

### Fingerprint

### ASJC Scopus subject areas

- Electrical and Electronic Engineering
- Condensed Matter Physics

### Cite this

**Measurement of rutile TiO2 dielectric tensor from 0.148 to 33 μm using generalized ellipsometry.** / Tiwald, Thomas E.; Schubert, Mathias.

Research output: Contribution to journal › Conference article

*Proceedings of SPIE - The International Society for Optical Engineering*, vol. 4103, pp. 19-29.

}

TY - JOUR

T1 - Measurement of rutile TiO2 dielectric tensor from 0.148 to 33 μm using generalized ellipsometry

AU - Tiwald, Thomas E.

AU - Schubert, Mathias

PY - 2000/1/1

Y1 - 2000/1/1

N2 - We have determined the complex uniaxial dielectric tensor of bulk rutile titanium dioxide (110), (100) and (111) samples using reflection generalized ellipsometry, which measures both the diagonal and off-diagonal elements of the reflection Jones matrix. Data were acquired using three commercially available ellipsometers, each covering the following spectral ranges: 0.148 to 0.292, 0.200 to 1.7 μm; and 1.7 to 33 μm. Generalized ellipsometry measures three complex ratios involving all four Jones matrix elements. In principle, this means that the complex dielectric tensor of a uniaxial crystal can be determined in a single measurement, provided that the sample is oriented such that the off-diagonal components of the Jones matrix are non-zero. To improve our results, we measure the samples at several rotational orientations around the surface normal. This insures that the probing electric fields vibrate along substantially different directions with respect to the optic axis. In some cases, we also varied the angle of incidence. The dielectric tensor was determined at every wavelength directly from a simultaneous fit to data from all rotational orientations and incident angles. A similar methodology should be applicable to a wide range of anisotropic optical materials.

AB - We have determined the complex uniaxial dielectric tensor of bulk rutile titanium dioxide (110), (100) and (111) samples using reflection generalized ellipsometry, which measures both the diagonal and off-diagonal elements of the reflection Jones matrix. Data were acquired using three commercially available ellipsometers, each covering the following spectral ranges: 0.148 to 0.292, 0.200 to 1.7 μm; and 1.7 to 33 μm. Generalized ellipsometry measures three complex ratios involving all four Jones matrix elements. In principle, this means that the complex dielectric tensor of a uniaxial crystal can be determined in a single measurement, provided that the sample is oriented such that the off-diagonal components of the Jones matrix are non-zero. To improve our results, we measure the samples at several rotational orientations around the surface normal. This insures that the probing electric fields vibrate along substantially different directions with respect to the optic axis. In some cases, we also varied the angle of incidence. The dielectric tensor was determined at every wavelength directly from a simultaneous fit to data from all rotational orientations and incident angles. A similar methodology should be applicable to a wide range of anisotropic optical materials.

UR - http://www.scopus.com/inward/record.url?scp=0033645688&partnerID=8YFLogxK

UR - http://www.scopus.com/inward/citedby.url?scp=0033645688&partnerID=8YFLogxK

M3 - Conference article

VL - 4103

SP - 19

EP - 29

JO - Proceedings of SPIE - The International Society for Optical Engineering

JF - Proceedings of SPIE - The International Society for Optical Engineering

SN - 0277-786X

ER -